Calibration is the comparison between measurement results of a device (unit under test) and another (standard) device with known or assigned values. Thus calibration only means recognizing the differences, whilst adjustment also means taking the necessary corrective actions to bring the unit under test into a defined tolerance.
Calibration should not be confused with the instruments white & dark calibration. Dark and white calibration of an instrument is performed on a daily basis by the user, utilizing the zero calibration box and the white calibration plate. These two calibrations are necessary for the instrument to work within a defined reflectance range. However, the white calibration plate may also change over time and therefore a real calibration service should be carried out once a year by an authorised service station.
While calibration only covers the recognition of the instrument’s condition, a maintenance service will assure that these conditions are within the defined specifications. Maintenance includes cleaning of the instrument, preventative maintenance and replacement of parts such as the xenon lamps as well as any re-adjustments necessary should the instrument not reach the specified accuracy. Every maintenance service also includes the calibration of the instrument.
Deciding on an interval for inspection or calibration depend on several factors. Some instruments are used in a laboratory or clean room environment which is mostly temperature controlled. Other instruments are used for quality control in a production line with harsher environmental conditions such as dust, humidity and a high number of measurements every day.
Konica Minolta recommends a calibration and maintenance interval of 1 year, which has been found to be suitable for most applications.
Procedures and tools for checking measuring instruments are strongly dependent on the type of instrument. As Konica Minolta has measuring instruments for a large range of different applications, the range of tools to be used is also very wide.
Konica Minolta utilises multiple standards for light and colour as well as for size. Of course, all of the standards used are traceable to national or international higher level standards. Details can be found in the traceability area where traceability charts for each kind of instrument are also available.
Light measuring instruments such as Luminance meters, Illuminance meters, incident colour meters and spectroradiometers are adjusted on an optical bench. It is a 5 meter long bench installed in a dark room. A certified tungsten lamp is controlled by a DC power supply and used as standard. By changing the measurement distance, different illuminance levels can be simulated. Certified diffusers are used as standard for luminance measurements. For spectroradiometers such as the CS-2000, the bench is used for level calibration whilst wavelength calibration is done by spectroscope.
Konica Minolta Sensing offer global service coverage for our measuring
instruments with a range of service and calibration options.